Class I Medical Device Recall: Medtronic SynchroMed II and SynchroMed EL Implantable Drug Infusion Pumps ? Feed Through Failure
Posted Jun 28 2013 12:00am
There is a potential for electrical shorting, internal to the SynchroMed infusion pump. An electrical short circuit in a feedthrough may present as a motor stall or low battery reset/alarm and may lead to a loss of or reduction in therapy.